SH-300 is equipped with a large 10.4" full-color display. The high resolution display provides for a flexible graphics menu combining respiratory curves, loops, and numerical data of physiological parameters.
Convenience for taking X-ray picture of the patient during ventilation, availability for clinician to assess patient's static pulmonary mechanics.
Accessibility for measuring intrinsic PEEP
More effective support for suction
The manifold parameter selections are operated by means of a single ComWheel
Availability for clinician to make prompt response to patient's inspiration need
With heating function, the expiratory valve protects the device from condensed water, accuracy of the flow sensor is ensure.
Warm and humidify airflow with protection function, nine steps to control temperature.
High quality medical compressor with low noise, compatible for most medical devices.
In VCV mode, the ventilator delivers mandatory breath by setting tidal volume. When the ventilator detects patient's inspiratory effort, it delivers a patient-initiated mandatory (PIM) breathe. If the ventilator does not detect inspiratory effort, it delivers a ventilator-initiated mandatory (VIM) breath at an interval based on the set respiratory rate. Breaths can be pressure-triggered or flow-triggered in VCV mode.
In PSV mode, inspiration is usually initiated by patient effort. Breaths are initiated via pressure or flow triggering. Clinician can also initiate a manual inspiration during PSV.
Controlled by pressure (preset PSV level + PEEP), Limited by pressure (preset PSV level + PEEP + margin), and Cycled by time (PSG Tmax) or flow (PSV Cycle)
SIMV is a mixed ventilation mode that allows both mandatory and spontaneous breaths. The mandatory breaths can be volume-based (SIMV-V) or pressure-based (SIMV-P), and the spontaneous breaths can be pressure-assisted. Clinician can select pressure-triggering or flow-triggering in SIMV.
APRV is a time cycled pressure mode. In this mode,
patient is allowed to breath spontaneously at two preset pressure
levels. These are set using the Phigh and Plow controls. The maximum
duration at each pressure during time cycling is set with the Thigh and
Tlow controls. Additionally, CPAP can be added to improve comfort of
patient during spontaneous breathing.