ENDO Hemodialysis Machine HD-1 is a machine used for purifying the blood of a person whose kidneys are not working normally.
When machine functions abnormally or electricity is cut, venous clamp responses automatically into close condition and stop blood transfusion in order to guard patient from continuously losing blood in extracorporeal circuit.
After blood is driven from patient’s body by blood pump, blood detection components detect and automatically make machine run into treatment mode. While after blood return back into patient’s body, blood detection components check the tubing. If there is without blood in the tubing, then response and stop the blood pump.
Air bubble detector can continuously check and catch single air bubble size bigger than 200μL while blood flow rate is 200ml/min. If there is air bubble bigger than 200μL in blood flow, machine alarms to keep patient from risk of air embolism. If accumulated air is excessive than certain volume in venous drip chamber, then trigger blood level detector, machine alarms, and stops blood pump and UF pump.
Easy to be found and seen with distance to shorten the time of solving problem.
850x470x1400 mm (W x L x H)
AC220V±10%, AC110V (optional) 50HZ
Pressure range:0.1-0.5 Mpa, Temperature range: 5-30℃
last for at least 30 min after electricity cut
300～800 mL/min linearity adjustable, Accuracy:-5%～10% mL/min
12-15.5 ms/cm, Accuracy: ±0.2 ms/cm
0～2000 mL/h, Accuracy:±30 mL/h (±5%)
30～500 mL/min, Accuracy:±10 mL/min
1～10 mL/h, Accuracy:±0.2 mL/h
2.5ml/min (syringe 10ml), 5ml/min (syringe 20ml), 7.5ml/min (syringe 30ml)
-300～+400 mmhg, Accuracy:±10 mmhg
-50～+300 mmhg, Accuracy:±10 mmhg
-100～+600 mmhg, Accuracy:±20 mmhg
single air bubble of 200μL exists when blood flow rate is 200ml/min
Chemical disinfection (citric acid, peracetic acid, sodium hypochlorite)