Color Graphic User Interface: Wide TFT LCD mount, clean and sharper image quality, unlike the existing electricity and the innovative new UI configuration operation.
Best Safety with Feedback System: REM (Return Electrode Monitoring) function monitors the size of contact area between a patient and the pad. If the size is inappropriate, it automatically blocks the high-frequency current to minimize the danger of burning incidents.
All Touch Screen System & Key: Touch screen system through the implementation easier and faster with a parameter setting, external key management can be the same as a touch screen.
Diversity of Feature Bipolar Bipolar Cut, Bipolar Blend, Bipolar Forced to add features to implement more up-grade the optimal system.
Option 1: Endo Cut (Intra cut) is a monopolar electrosurgical treatment method and a cut which is carried out intervals (fractionated cut) with software-controlled alterations between cutting and coagulation stages.
Option 2: T.U.R.P. is a special resectoscopy, which removes the tissue in the prostate with kinetic plasma under saline liquid. Vessel Sealing up to 7mm with bipolar coagulation during the procedure.
10 storage are available by using user storage function.
2 (two) monopolar handpiece can be connected to the device.
Bipolar cut function is available.
A display window allows you to easily check output value, indicated as digits.
Unit provides various functions that are required to perform an operation as functional. Cut mode (Pure cut, Blend1, Blend2, Blend3), coagulation Mode (Spray coagulation, Contact coagulation), bipolar coagulation mode by hand or foot, bipolar cut (only available in ZEUS upgraded version)
Each working condition can be checked by sound and indication lamp.
Hand bipolar (Auto bipolar) can be checked only by contacting forceps.
Unit output strong and stable power by Micro-processor.
Input power frequency is automatically detected.
A fuse which is built in a power circuit board prevents over current from flowing through the equipment.
When the plate attached to the patient is disconnected, red alarm light begins flickering.
REM (Return Electrode Monitoring) function monitors the size of contacting area between a patient and the pad. If size is inappropriate, it automatically blocks the high-frequency current to minimize the danger of burning incident.
To protect a patient, the case is fully grounded so that a leakage current can flow into the earth.